USER GUIDE Model F-805
INSTRUCTIONS FOR THE F-805 Galvanic Machine For Single or Double Roller Version Galvanic Machines
Instructions for treatment . Theory Galvanic current is a direct current, which causes a chemical change in the tissues by reacting with the sodium chloride salts in the body. Galvanism uses two electrodes both applied to the client’s body, one strapped on or held by the client and the other in contact with the client by means of the therapist moving it across the skin. The electrodes have a specific polarity and a switch of a button on the machine can change that polarity. Within the galvanic treatment the electrodes act as poles, working to attract or repel the actively charged elements in the desincrustation or iontophoresis phase. General effects being deep cleansing and hydrating.
. General Usage NB: - If you have a twin roller machine with no earth bar, the black lead roller acts as your earth/indifferent bar (there is no need for the client to hold the earth bar, as current travels between the rollers) to maintain the circuit continuity. (1). Connect all the machine accessories before turning on the power. The Client should be holding the earth bar, prior to turning the power on and the therapist using the rollers or desincrustation attachments. (2). To change the polarity of the rollers, use the polarity switch to (+R) for postive current, or to (-G) for negative current. The skin should be cleansed prior to treatment to ensure the area is clean, which will increase the benefits of the galvanic treatment. Turn the electrode button to (+R), now the roller is positive while the indifferent bar (earth bar) is negative, or visa-versa. Then, using the desired product of your choice set the machine to the correct polarity for the skincare products and carry out the treatment phase according to your product brand training schedule. Generous application of the product will always ensure a good conductive circuit is made and maximum treatment results are achieved for your client (3). Power on the “IONOS”, and select the polarity setting and then turn the “I/Energy” to the minimum setting. You may then make contact on the skin with the rollers and start your treatment phase. You may then, as required, turn up the output slowly by adjusting the “I/Energy” to the appropriate intensity at which the client feels a mild tingling sensation. Once the client feels the tingling sensation, you should leave the machine setting at this level for the duration of the treatment. You may then smoothly slide the rollers over the skin, in the movements laid out by the skincare product supplier.
Caution and Contra-Indications NB: -After treatment, the rollers and/or indifferent/ earth pole must be cleaned (or sterilized) and dried thoroughly before its next use. Due care should be taken during cleaning or sterilizing of the attachments. This should be observed to prevent damage or a shortening of the attachments lifespan. (1) Some clients may feel a strong, or painful tingling sensation during initial phase of the treatment, which may be due to hypersensitivity to the current, or the machine output setting is incorrect. The output should be reduced immediately if the client feels any discomfort or more than a mild tingling sensation. Failure to reduce the output level or holding the rollers in one position may result in galvanic burns. (2) Prior to any electrical treatment any jewellry (including body piercing’s) must be removed. (3) Suitable galvanic (polarized) gels must be used with the machine at all times to achieve maximum results for your client, and maximum performance from the your machine. (4) In the course of using the roller/s, they should not be touched together, if any accidental contact between the rollers/earth bar happens, the instrument safety system will stop the current output to protect the client automatically. At this point, you will need to turn the “I/Energy” to its minimum position and restart it again, by turning off and then on, the power switch. You will then need to reset the out put you need too. (5) Each bar (LED) on the power display represents 0.1 of a Milliamp. The machine runs at a standard facial output of 0.8 to 0.85 milliamps. Should your client require, or the treatment you use require a stronger output, you may adjust the output setting to a maximum of 1.25 ma when all LED’s are lit. Should you try to increase the out put further than this, the safety cut out will engage and the machine will require restarting (as per step 4 above). (6) When powering off the instrument, ensure you turn off the “I/Energy” first, then the “IONOS” switch, to prevent any damage to your machine. (7) If only the power indicator lights when switching on the instrument, the “I/Energy” is in the stand-by status, but if the energy output lamp has no indication, you should switch off the “I/Energy” and restart it. (8) General contra-indications to electrical facial treatments: Allergy to metal/nickel Cancer Metal pins/plates Pregnancy Skin diseases Silicone implants Roaccutane (within last 12 months) Epilepsy Pacemaker Thrombosis/phlebitis Operations (within last 6 months) Hypersensitive/vascular skins
F-805 Overview Power level indicator Bar
Before switching on the machine, ensure the power output dial is turned fully to the left. The LED power indicator shows increments of 0.1 ma (milli-amp). The machine is preset to run at .8ma output for standard facial procedures. Should you wish to raise the output, the machine will rise to 1.25ma when all LEDâ€™s are lit. If the dial is turned pass the 1.25ma (max) setting, the auto safety cut out, will Switch the power output off automatically for safety, to prevent galvanic burns. The machine will require the output dial fully to the left, to restart after saftey mode. The machine should only be switched to the higher settings if the client is intolerant to current, or you are conducting treatments such as body desincrustation etc.
Machine will cut out when Maximum output is reached.
Polarity Switch + = Red LED - = Green LED
Power Level Dial
Power On/Off Switch
Warranty Procedure 1 - The warranty on all Cosmetronic equipment is 12 months from the date of invoice and covers the products against defects in materials and workmanship or component defect. (Unless superseded by an extended warranty agreement) This warranty is non transferable and applies to the original purchaser only. 2 - No product repairs or component replacement will take place unless all/any amounts owed to Cosmetronic for the said goods have been paid for in full, prior to any repairs. 3 - All extended warranty policies must be paid in full prior to dispatch of goods and will extend the above cover (No 1) by a further 2 or 4 years, to a maximum of 5 years including the first free year. 4 - Before returning any product to Cosmetronic UK Ltd for repair or replacement you must first: - Endeavour to establish cause of product fault (e.g.. check fuses etc.). - Notify Cosmetronic of any fault development within 24 hrs of its occurrence. - Contact a member of our technical service dept (95% of faults can be rectified over the telephone). - Obtain an RMA (return material authorisation) from a member of Cosmetronic. - Package the goods to include all parts or accessories received with that item. - Any item received incomplete for repair will be made complete and all relevant parts charges added to repair. - Pack the goods to be returned in sufficient packaging as to prevent any damage occurring in transit, as Cosmetronic will not be held liable for damage to products due to insufficient packaging. - Clearly address the returned goods with our full postal address and RMA code. 5 - All goods returned to Cosmetronic with out the correct RMA authorisation will be refused. 6 - No goods will automatically be replaced, when the faulty item can be repaired within a reasonable time span (except items deemed unserviceable on arrival). Goods will be replaced at the discretion of Cosmetronic when it is deemed that all/any goods are beyond economic repair. Items of loan equipment may be given at the discretion of the management and subject to delivery/collection charges, together with a refundable deposit. 7 - Cosmetronic will not be held liable for any momentary lack of stock or technical resource, or any other circumstance beyond its control that may delay any component replacement or repair. Cosmetronic will not be held liable for any special, incidental, consequential damages resulting from a breach of warranty, or under any other legal theory, including but not limited to loss of profit or good will, downtime, damage to or any products, treatments used in conjunction with its products. Purchase implies your acceptance of Cosmetronic terms & conditions. 8 - When our service department cannot determine any fault or malfunction with the goods returned, you may be liable for any administration and testing costs. 8a- When goods returned are not in a merchantable (dirty or unhygienic) condition, they will be made to merchantable quality again, and cleaning costs will be charged where necessary. 9 - All goods returned will be at the owners risk. Cosmetronic will not be held liable for any damage or loss to goods whilst in transit. 10 - Carriage costs are at the owners expense for both collection & return freight costs. All goods will be carried via Cosmetronic nominated couriers unless specified. Any goods in transit insurance required must be requested prior to collection/dispatch. 11 - No repaired goods will be released until any costs relevant to that repair (unless under warranty) have been settled in full, this will include all carriage/freight costs. 12 - Call out/on-site repairs will be carried out as soon as is physically possible and charged at a cost of ÂŁ POA. The charge is based on the original delivery address of the product when purchased. This charge is payable in full, prior to any on site attendance. This charge will not include travel expenses (fuel, parking, toll charges etc.) but, will include labour and the parts replacement of electrical or motorised mechanical components, to include electrical motors, foot control boxes, system control boxes but does not include optional extras such as Vibro or Thermo heat functions. The call out repair charge will not cover items deemed as natural â€œwear & tearâ€? such as upholstery etc. The call out charge will not cover damage by accident, misuse, or misapplication, or if the product has been modified without permission. Any serial numbers removed or displaced will also result in warranty invalidation. Proof of purchase will be required before any repairs carried out. 13 - Warranty call out, or on-site repairs, will be charged as per the prevailing rate (POA), terms and conditions, but will be refunded once established that any fault is covered under and within the warranty period and criteria.